I was using the load file. Message 9 of We are using the device Keithley and yes i have two different temperatures i differentiated one with hot water. I don’t understand what did i do wrong. Message 29 of It’s the same problem as described in here:
|Date Added:||23 April 2015|
|File Size:||38.39 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Most Active Software Boards: You should see at least wo entries if all is working:.
Auto-suggest helps you quickly labveiw down your search results by suggesting possible matches as you type. Once you have determined the model, ensure you have installed the correct and latest driver.
Message 28 of My best guess would be that there is some sort of race condition where the instrument is not processing calls from the instrument driver in the correct sequence.
Message 25 of What was the driver that you used in conjunction with the rest of the code? Do you have one thermocouple very warm and the other very cold? Message 30 of Thanks again to all. Hi Jesse, I couldn’t find a manual for thebut I found one for the Keithley I also had luck changing the aperature timing by placing it after the config measurement.
Keithley Multimeter/Data Acquisition Sy – Instrument Driver – National Instruments
In order to resolve this issue, a “Decimate 1D Array” function is used in order to separate values from individual channels into separate rows or columns in an array.
Labviea Active Software Boards: I was using the load file. How are you verifying that the temperature readings are switching over?
It often occur at some intervals and intervals doesn’t seem continous. Message 26 of Auto-suggest helps you quickly narrow down your search results by suggesting labvew matches as you type.
If you have the NI I don’t understand what did i do wrong. Got the user manual this morning and am reading so I can keithhley commands to the machine. Can you please look in to it.
I tried using it for the measurement of resistance ieithley seems like it gives an error. The reason for this is because you can configure several different measurements on the same scan using several scan lists and several config meas.
I put the GPIB Thanks for the VI. To test VISA communication: I am not familiar with the express vi “Write to Measurement File”, but you could separate yours data like this:.
keithley driver problems – Discussion Forums – National Instruments
I have attached a picture of the current scan setup to the loop. In order to separate data from these two channels into their own arrays, the “Decimate 1D Array” is necessary. We are using the device Keithley and yes i have two different temperatures i differentiated one with hot water. Seems you are also confused about the device The Keithley has a configurable scan rate of Fast, Medium, and Slow.